/*Dictionary of Terms*/
Test tape mode is now more tolerant of noise and glitches. FSK decoding can now operate on the R channel. Screen updates are now much faster during fast repeating sweeps. Crosstalk segments with measurement levels are now allowed. Sequence error 13 is generated if a noise or crosstalk segment is received without a test level segment. Input impedance is no longer printed with sequence results since it made the printout look cluttered. A Test Level result in dBu indicates that 10k was used, while dBm indicates that 600 was used.
MC and HC impedance corrections have been removed from manual mode because they were too confusing and of very limited use, although they are still available via remote mode (section 9.8) and sequence mode.
Matched Correction at 600 is the same as 600 ZC, and is of little use with 10 or 75 output impedances since the LA101 cannot drive a 10 or 75 load at a very high level.