These sheets are intended to supplement the manual.
Download Info Sheet for v3.05 - PDF 20k
Download Info Sheet for v2.98 - PDF 20k
Download Info Sheet for v2.8 - PDF 20k
Download Info Sheet for v2.4 - PDF 20k
Download Info Sheet for v2.3 - PDF 20k
Standard Sequences (V3.05 firmware chip)
MS1 Units have SEQ1 only, but can be upgraded.
SEQ0 only runs at initial switch-on
To run SEQ5 hold the SEQ key down and press key 1
To run SEQ6 hold the SEQ key down and press key 2 (etc to SEQ 8)
To run SEQ9 hold the SEQ key down and press LEV
To run SEQ10 hold the SEQ key down and press NOISE
To run SEQ11 hold the SEQ key down and press DIST
To run SEQ12 hold the SEQ key down and press PPM
To run SEQ13 hold the SEQ key down while pressing dBu/REL
NB: Because sequence results are held until cleared by the running of a master-seg (n or m) or by resetting, extra segments can be run to add in to results already displayed in Lin4Win. For example, it can be useful to run SEQ3 and then run SEQ5 (slow sweep) or SEQ8 (wideband sweep), replacing the original sweep. Or to run SEQ3 and then SEQ10 (noise plot), which will add to the results. Running SEQ13 will add crosstalk results.
Segments available (V2.98 firmware)
Any combination of segments (up to 6) can be sent as a sequence using the Lin4Win support software running on a PC, with the following provisos:
Only one ‘MasterSeg’ can be sent (n or m)
Only one Sweep segment can be sent (u, or s, or v, or w, or h)
Only one extra plot can be sent (l or z)
Seg z only works with a sweep (it only signals the showing of phase)
Segment letters can be entered in any order, either in the control panel or at the top of the main (results) window (press return to send) – the sending order is fixed (eg nurqcl)